Dissolved oxygen (DO) is the free oxygen available in solutions and is consumed in the metabolism of all aerobic organisms. DO concentration in water is used as an indicator of the health of ecosystems, and it varies with solution temperature, agitation, salinity, injection rate, and barometric pressure (Al-Rawahy et al., 2019; Wei et al., 2019). High temperatures and salinities decrease the capacity for DO. Most DO meters have temperature ranges of 0 to 40 °C and salinity ranges of 0 to 70 g⋅L–1 (typically from NaCl). These ranges encompass environments ranging from arctic to tropical and from freshwater to seawater.
Demand for DO in respiration increases with biological density, metabolic rate, growth stage, and temperature. High biological demand requires greater inputs of oxygen to sustain homeostasis (Ben-Noah and Friedman, 2018). The nutrient solution of liquid hydroponics benefits from levels of DO close to saturation near 8 mg⋅L–1. This promotes healthy root respiration and minimizes hypoxic microsites or the development of anaerobic bacteria (Schroeder and Lieth, 2004). DO concentrations less than 5 mg⋅L–1 can lead to reduced root respiration and can result in the development of adventitious roots (Holtman et al., 2014). DO is necessary for nitrification in aquaponic systems; concentrations near saturation promote aerobic bacteria and nitrification whereas depleted levels promote anaerobic bacteria and denitrification (Wongkiew et al., 2017; Zhen et al., 2015). Accurate measurement of DO helps identify hypoxic zones even in seemingly well-aerated solutions.
DO can be measured using three methods: iodometric titration, membrane diffusion (electrochemical), and fluorescence quenching (optical) (Tai et al., 2011; Zaitsev et al., 2018). Iodometric titration is a colorimetric titration and is labor intensive. More than eight companies sell a DO meter that uses optical technology, and more than 12 companies sell a meter with electrochemical technology. The meters tested in our review represent about 20% of the electrochemical DO meters available. Meters using electrochemical technology are less expensive than optical meters. With electrochemical meters, DO crosses a semipermeable membrane, is reduced at the cathode, and produces a voltage difference with the anode that is proportional to oxygen concentration (Parra et al., 2018).
There are two types of electrochemical technology. Polarographic sensors have a gold cathode and silver anode that are polarized temporarily at 0.8 V during measurement. Galvanic sensors have a silver cathode and zinc anode that are polarized constantly even when the meter is powered off (Wei et al., 2019). Polarographic electrodes require a warmup period of up to 15 min. Galvanic electrodes do not require a warmup period, but constant polarization decreases the lifetime of the electrode.
Sensor technology determines price. Galvanic sensors are typically less expensive than polarographic sensors, which are less expensive than optical sensors. Optical sensors typically have a faster response and greater resolution than polarographic and galvanic sensors. We evaluated DO meters at three price points for the accuracy, stability, and response time across salinity and temperature.
Al-Rawahy, M.S., Al-Rawahy, S.A., Al-Mulla, Y.A. & Nadaf, S.K. 2019 Influence of nutrient solution temperature on its oxygen level and growth, yield and quality of hydroponic cucumber J. Agr. Sci. 11 75 92
Bar-Yosef, B. & Lieth, J.H. 2013 Effects of oxygen concentration in solution and uptake rate by roots on cut roses yield, and nutrients and sugars content in leaves Scientia Hort. 155 49 55
Ben-Noah, I. & Friedman, S.P. 2018 Review and evaluation of root respiration and of natural and agricultural processes of soil aeration Vadose Zone J. 17 1 47
Goto, E., Both, A.J., Albright, L.D., Langhans, R.W. & Leed, A.R. 1996 Effect of dissolved oxygen concentration on lettuce growth in floating hydroponics Acta Hort. 440 205 210
Holtman, W.L., Oppedijk, B.J., Vennik, M. & van Duijn, B. 2014 Low oxygen stress in horticultural practice 381 394 van Dongen, J. & Licausi, F. Low-oxygen stress in plants. Plant Cell Monographs
Langenfeld, N. & Bugbee, B. 2021 Dissipation rates of oxygen nanobubbles in recirculating systems Utah State Univ. Crop Physiol. Lab Techniques Instrum. 19 1 3
Schroeder, F.G. & Lieth, J.H. 2004 Gas composition and oxygen supply in the root environment of substrates in closed hydroponic systems Acta Hort. 644 299 305
Tai, H., Yang, Y. & Liu, S. 2011 A review of measurement methods of dissolved oxygen in water Proc. Intl. Conf. Computer Comput. Technol. 2 569 576
Wei, Y., Jiao, Y., An, D., Li, D., Li, W. & Wei, Q. 2019 Review of dissolved oxygen detection technology: From laboratory analysis to online intelligent detection Sensors (Basel) 19 3995 4033
Wen, Z. & Zhong, J. 1995 A simple and modified manometric method for measuring oxygen uptake rate of plant cells in flask cultures Biotechnol. Tech. 9 521 526
Wongkiew, S., Hu, Z., Chandran, K., Lee, J.W. & Khanal, S.K. 2017 Nitrogen transformations in aquaponics systems: A review Agr. Eng. 76 9 19
Zaitsev, N.K., Dvorkin, V.I., Melnikov, P.V. & Kozhukhova, A.E. 2018 A dissolved oxygen analyzer with an optical sensor J. Anal. Chem. 73 102 108
Zhen, H., Lee, J.W., Chandran, K., Kim, S., Brotto, A.C. & Khanal, S.K. 2015 Effect of plant species on nitrogen recovery in aquaponics Bioresour. Technol. 188 92 98